Within the framework of a collaborative project between the DES/DDSD and the DRF/IRFU, a feasibility study of muography potential for the auscultation of nuclear reactors was initiated in 2017. After an initial evaluation phase carried out by IRFU using numerical modelling, first data were taken on the G2 reactor block, located at CEA Marcoule and shut down in the early 1980s, from February 2020. These measurements demonstrated the potential of the technique, identifying differences between the current structure of the G2 reactor and the 3D model created from the original plans of the installation. These initial results demonstrate the interest of using muography in the clean-up and dismantling of nuclear facilities, one of the CEA's priorities nowadays. For the next phase of the project, a 3D tomography of the reactor is envisaged by combining images taken from different positions. It could be the first 3D image of the interior of a reactor at dismantling phase without using any artificial ionizing radiation. This will provide a new inspection tool to the existing palette.
In order for the images produced by the future MRI to be free of distortions or artifacts, the magnetic field generated by the Iseult magnet must be homogeneous to 0.5 PPM (parts per million) around the patient's brain. To meet this challenging specification, it was necessary to provision means of "shimming" the field, i.e. of correcting all the small defects that would inevitably arise from the manufacturing process. 5904 pieces of shim (small iron platelets) were screwed onto rails and installed inside the magnet tunnel. This first configuration was tested on Thursday, July 9, 2020 by mapping its effect on the magnetic field of Iseult at 3 T. The results are very encouraging as this first shimming iteration allowed to increase the homogeneity of the field in the useful zone from 138.8 to 3.2 PPM (value extrapolated to 11.72 T from magnetic measurements at 3 T).
In its most common version, muon imaging is intrinsically a 2D technique: the resulting density map is indeed integrated along the observation direction. However, a 3D map can be obtained by combining several projections, like for medical imaging. But in the muon case, the number of projections is dramatically reduced because of the required acquisition time. A 3D algorithm has been recently developed using the Irfu TomoMu setup, within a collaboration between Florence University and Irfu. The 3D structure of the test object has been reconstructed from only 3 points of view, thanks to the high precision of the instrument. This technique will soon be extended to more applications, from reactor dismantling to civil engineering or mining exploration.